Jet valve driver zc1
[available since BMT Software version 0.224]
Read the section about Adding a new AddOn Device.
Each computer needs to be licensed for usage of the ZC1 controller with the software.
|Peak current (Ip)||Peak power value in 50 mA steps from 200 mA to 1400 mA (default 1200 mA)|
|Holding current (Ih)||Holding power value in 50 mA steps from 50 mA to 722 mA (default 274 mA)|
|Release mode||Operating mode selection.
Software = Software control
|Peak time (tp)||Peak time in [μs]
Value between 100 and 4.000
|Valve opening time (to)||Opening time in [μs] |
If Open-time < peak-time, the peak time is cutted down, ie Peak time = Open time.
|Cycle time (T)||Cycle time in [μs]|
If cycle-time < open-time, the value is automatically adjusted. Cycle-time=open-time + 100 μs
|Frequency (Hz)||Frequency of dosing interval.|
Calculated according to valve opening time (to) and cycle time (T)
|On||Opens the valve|
|One shot||One shot with duration of valve opening time (to)|
|Shot series||Number of shot series using control parameters|
|Endless shot series||Endless shots using control parameters|
|single||The functions of the program table will be run once.|
|repeat||The functions of the program table will be repeated.|
|repeat endless||The repeating is endless and will only be stopped when the exception is stopped using the program stop button in the controller or program details.|
|repeat time||The program table will be repeated for the time entered in this field.|
|Program table||This table shows the configuration of the different program steps.|
It is possible to re-order or delete the entries using the buttons at the right of the table, or the table menu using the right mouse klick.
Action = Different program action (Single shot, Shot Series, On, Endless shot series, Stop, Sleep)
Control Parameter = Setting new control parameter (Peak time, Valve opening time, Cycle time, Frequency)