General Safety Instructions
For Research Use Only (RUO) - Laboratory Instrument
General Safety Instructions
The warning and caution symbols are described below. Please make sure the operators fully understand the meanings of the symbols to help to ensure your personal safety and to help to protect your equipment and working environment from potential damage.
Only use the device with the included power supply XP Power Model ACM36US24 (Output 24VDC 1.5A).
|This symbol indicates matters that may lead to imminent risk of death or serious injury if ignored or incorrectly handled.|
|CAUTION: TO PREVENT ELECTRICAL SHOCK, DO NOT REMOVE ANY COVER SCREWS.|
|Install in accordance with the supplied user manual.|
|Unplug this unit during lightning storms or when unused for a long period of time.|
|Do not attempt to disassemble, repair or modify the system. Failure to do so may cause electric shock or injury.|
|Do not cover or block any ventilation openings.|
|Do not drop, nor give strong shock and vibration to the system. Failure to do so may cause malfunction.|
|Unplug the system or turn the power off in case of abnormality. Failure to do so may cause fire and/or electric shock.|
|Avoid wet locations, or locations with humidity, dust and/or oily smock, or near a heat generating device. Failure to do so may cause malfunction and/or electric shock.|
|Protect the power cord from being walked on or pinched, particularly at plugs, and the point where it exits the unit. Do not place heavy items on the power cord, which may damage it. Take note that the power cord cannot be reached and pulled by young children, avoiding injury.|
Where to install Place the unit on a sturdy platform of which the surface should be flat. This will prevent it from falling over. Make sure the unit is located in a position where it cannot be pushed or hit by objects, as pressure will break or damage the unit, and so that small objects cannot be inserted into the ventilation slots or openings in the cabinet.
|This symbol indicates that the relevant electrical product or battery should not be disposed of as general household waste in Europe. To ensure the correct waste treatment of the product and battery, please dispose them in accordance to any applicable local laws of requirement for disposal of electrical equipment or batteries. In so doing, you will help to conserve natural resources and improve standards of environmental protection in treatment and disposal of electrical waste (Waste Electrical and Electronic Equipment Directive).
When you see a crossed-out wheeled waste container on the product. It means that the product falls under the EU directive 2002/96/EC
Users Qualification and Operation conditions
Always make sure that only appropriately trained and qualified persons are working with the controller and the connected fluidic components (pumps & valves) and always supervise their operation. The work on the electrical components and the controller systems must be done by persons with the required qualifications and expert knowledge.
Basically, BMT Fluid Control Solutions GmbH cannot be held responsible for any improper or undue liability negligent use. This is especially valid for the operation of pumps, valves and other components not certified by BMT Fluid Control Solutions GmbH.
Do not open the housing of the controller! In such cases, BMT Fluid Control Solutions GmbH can no longer guarantee.
Do not make any modifications or additions to the controllers without prior written consent of the manufacturer. Such modifications may affect the safety of the system or the connected fluidic components and are therefore prohibited. BMT Fluid Control Solutions GmbH rejects any responsibility for the damage of systems caused by unauthorized modifications of the controllers or the fluidic components. The risk and responsibility are thus automatically transferred to the operator.
Please keep this manual in a safe place and give each user access to a copy.
Piezo pump controller
Please note that parts of the electronics and the piezo pump are operated with high voltages. Therefore, people with pacemakers are advised to avoid the proximity to the overall system.